---
product_id: 135070800
title: "HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug"
brand: "hiletgo"
price: "E£ 3701"
currency: EGP
in_stock: true
reviews_count: 5
category: "Hi Letgo"
url: https://www.desertcart.com.eg/products/135070800-hiletgo-usb-logic-analyzer-device-with-emi-ferrite-ring-usb
store_origin: EG
region: Egypt
---

# 24MHz max sampling rate ±42ns pulse-width accuracy 8 versatile input channels HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug

**Brand:** hiletgo
**Price:** E£ 3701
**Availability:** ✅ In Stock

## Summary

> ⚡ Unlock pro-level debugging with 24MHz speed and 8 channels of pure insight!

## Quick Answers

- **What is this?** HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug by hiletgo
- **How much does it cost?** E£ 3701 with free shipping
- **Is it available?** Yes, in stock and ready to ship
- **Where can I buy it?** [www.desertcart.com.eg](https://www.desertcart.com.eg/products/135070800-hiletgo-usb-logic-analyzer-device-with-emi-ferrite-ring-usb)

## Best For

- hiletgo enthusiasts

## Why This Product

- Trusted hiletgo brand quality
- Free international shipping included
- Worldwide delivery with tracking
- 15-day hassle-free returns

## Key Features

- • **Crystal-Clear Accuracy:** ±42ns pulse-width measurement ensures reliable, professional-grade diagnostics.
- • **Plug & Play Simplicity:** Includes USB & Dupont cables for instant setup—get debugging without delay.
- • **Wide Voltage Compatibility:** Seamlessly works with 5V, 3.3V, 2.5V, and 2.0V systems for flexible integration.
- • **Blazing 24MHz Sampling Speed:** Capture ultra-fast signals with precision timing for next-level debugging.
- • **8-Channel Multi-Protocol Support:** Simultaneously analyze UART, IIC, SPI and more across 8 inputs.

## Overview

The HiLetgo USB Logic Analyzer is a compact, 8-channel device offering up to 24MHz sampling rate and ±42ns pulse-width accuracy. Compatible with multiple voltage levels and protocols like UART, IIC, and SPI, it includes essential cables for immediate use. Ideal for engineers and hobbyists seeking affordable, precise signal analysis.

## Description

HiLetgo USB Logic Analyzer Device With USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz; The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel; Voltage range: Input voltage range: -0.5V to 5.25V Input Low Voltage: -0.5V to 0.8V Input High Voltage: 2.0V to 5.25V Works with 5V, 3.3V, 2.5V, 2.0V systems. May work with 1.8V but not recommended. Input Impedance: 1Mohm || 10pF (typical, approximate) Crystal: +/-20ppm, 24MHz Error/Accuracy: pulse-width measurement: +/- 42ns (at 24MHz). Packing Included: 1 * Logic Analyser 1 * USB Cable 1 * Dupont Cable

Review: I attached this logic analyzer to an old modem's UART and was able to capture the modem's boot process using PulseView. The device was recognized as 'Saleae'. This is a great device for hobbyists: basic, simple and very affordable. I am happy with this purachase
Review: Good,It's working I doubted but no problem working fine.

## Features

- The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel
- Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
- The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
- Input voltage range: -0.5V to 5.25V; Input Low Voltage: -0.5V to 0.8V; Input High Voltage: 2.0V to 5.25V
- Input Impedance: 1Mohm || 10pF (typical, approximate); Crystal: +/-20ppm, 24MHz

## Technical Specifications

| Specification | Value |
|---------------|-------|
| ASIN | B077LSG5P2 |
| Batteries Required? | No |
| Batteries included? | No |
| Best Sellers Rank | 164,016 in Business, Industry & Science ( See Top 100 in Business, Industry & Science ) 204 in Lab Oscilloscopes |
| Colour | Gray |
| Customer Reviews | 4.5 4.5 out of 5 stars (531) |
| Date First Available | 19 Nov. 2017 |
| Item Package Quantity | 1 |
| Item Weight | 68 g |
| Item model number | 3-01-0015 |
| Manufacturer | HiLetgo |
| Package Dimensions | 19.3 x 8.89 x 1.4 cm; 68 g |
| Part number | 3-01-0015 |
| Shape | Round |
| Size | S |
| Special Features | Low Voltage |

## Product Details

- **Brand:** HiLetgo
- **Cable type:** USB
- **Compatible devices:** Modem
- **Connector type:** USB Type A
- **Special feature:** Low Voltage

## Images

![HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug - Image 1](https://m.media-amazon.com/images/I/619wsd01FbL.jpg)

## Available Options

This product comes in different **Size** options.

## Questions & Answers

**Q: What software is required for Mac or PC?**
A: https://support.saleae.com/hc/en-us/articles/115005972023-Saleae-Logic-Beta-Software
now with real time (ish) display

**Q: What protocols does this support?**
A: This product does not natively support any protocols. The protocol decoding is done by the host software. Check out the feature sets for PulseView  and Saleae Logic software.

**Q: Anyone get this error msg?  the application was unable to start correctly (0xc000007b). click ok to close the application.**
A: I was experiencing the same error with Sigrok's PulseView x64 and ZaDiag x64 not properly starting, with the OS tossing a 0xc000007b error dialog every time I tried to launch either app.

As with the other answerer, the 32bit version would install and would start, but not work correctly with this LA.  I ended up giggling my way to success in determining that I didn't have the right .net libs yet installed on this box (a fresh install of Win10 x64).  This site (www.techjunkie.com/0xc00007b/) pointed me to the right download needed, in the section 'HOW TO FIX ‘THE APPLICATION WAS UNABLE TO START CORRECTLY 0XC00007B’ ERRORS' - I did the Microsoft Visual C++ 'x64' recommendation (the 'x64 version' of the MS Visual C redistributable).  After this, I was able to install and successfully launch the x64 versions of both the PulseView and ZaDiag apps.

Currently, I am still struggling with getting PulseView x64 to work with this LA correctly, even after using ZaDiag x64 to change the driver to the default one it offers when it launches.   Still working that and will update once I get it.

For now, I reverted to using Saleae's Logic 1.2.29 Beta app (available from heresupport.saleae.com/logic-software/latest-beta-release - link good as of 10Jun19).

To get Saleae's app to work with it, I had to repeatedly uninstall the device (including deleting the drivers) from the MMC Device Manager snap-in until, when searching for new plug and play hardware, it shows up as 'unknown device'. Once to that point, running the Saleae app should result in it seeing the driverless device with the HW IDs it is looking for and will assimilate it and install the drivers for it, and instantiate it as a 'USB Serial Bus controller|Saleae Logic USB Logic Analyzer', enabling this LA to be used from the Saleae Logic 1.2.29 app (images: i.imgur.com/J60jm8A.png & https://i.imgur.com/NA0MabE.png).

As indicated above, once I get this LA to work with the 64-bit version of PulseView/ZaDiag properly, I will update this answer with additional details.  Until then, take care.

-t

**Q: Driver installation fails - don't know where to get support

There is no web site or email address. I am following instructions posted here.**
A: Are you using PulseView? FWIW It worked smoothly on my Mac...

## Customer Reviews

### ⭐⭐⭐⭐⭐ Review
*by F***B on 26 June 2022*

I attached this logic analyzer to an old modem's UART and was able to capture the modem's boot process using PulseView. The device was recognized as 'Saleae'. This is a great device for hobbyists: basic, simple and very affordable. I am happy with this purachase

### ⭐⭐⭐⭐⭐ Review
*by B***N on 3 January 2026*

Good,It's working I doubted but no problem working fine.

### ⭐⭐⭐⭐⭐ Review
*by C***R on 17 January 2019*

Summary Pros - inexpensive, works well with free open source software Cons - no on board capture buffer, no probes included This is a good little logic analyzer for the home hobbyist or even a small development team. The hardware uses a CY7C68013A microcontroller (an 8051 MCU core with an integrated USB 2.0 interface). It comes with a USB A to mini USB B cable and some short jumpers wires. It is powered by USB so there is no other power supply to deal with. There is no on board capture buffer so your computer needs to be fast enough to receive the data as it comes over USB. With most systems that shouldn’t be an issue providing you are not doing any other intensive tasks, particularly those that use USB, at the same time. Trying to run all 8 channels at 24Mhz though could be an issue on some systems. Looking at the board, there isn’t any input buffering other than series resistors. Care must be taken not to expose the channel inputs to voltages that could damage the on board microcontroller. There is no software included, but one can download the open source sigrok/PulseView which fully supports this device. I tested it with PulseView 0.4.1 on a Windows 10 system and it was recognized as “Saleae Logic” analyzer (an older name brand analyzer). The only part that could be slightly tricky to getting PulseView working is installing the WinUSB driver using Zadig. If you are unfamiliar with Zadig, you can read about it on the sigrok wiki area for windows. Since this board emulates a “Saleae Logic”, there are also commercial software offerings that should work with it. I did an initial test using two channels simultaneously reading a 115200 bps data stream (sample rate set to 500kHz) and it worked flawlessly. PulseView easily decoded the ASCII data once configured. I again captured the same stream setting at 24Mhz (which is way over sampled for the data stream) just to see what it would do, and it worked fine showing the much more frequent sample rate. I’m not aware of a way to setup anything but a simple trigger to start/stop capture using PulseView. That is a software limitation if you are trying to catch a more complex event that occurs infrequently. The jumper wires that came with it are of limited use unless you always have available header pins to connect to in your circuit (doubtful). I suggest ordering a set of “logic analyzer test clips” to complete your setup.

## Frequently Bought Together

- HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
- ELEGOO 120pcs Multicolored Dupont Wire 40pin Male to Female, 40pin Male to Male, 40pin Female to Female Breadboard Jumper Ribbon Cables Kit Compatible with Arduino Projects
- Goupchn SMD IC Test Hook Clip 12PCS 6 Colors for Logic Analyzer, Electronic Testing

---

## Why Shop on Desertcart?

- 🛒 **Trusted by 1.3+ Million Shoppers** — Serving international shoppers since 2016
- 🌍 **Shop Globally** — Access 737+ million products across 21 categories
- 💰 **No Hidden Fees** — All customs, duties, and taxes included in the price
- 🔄 **15-Day Free Returns** — Hassle-free returns (30 days for PRO members)
- 🔒 **Secure Payments** — Trusted payment options with buyer protection
- ⭐ **TrustPilot Rated 4.5/5** — Based on 8,000+ happy customer reviews

**Shop now:** [https://www.desertcart.com.eg/products/135070800-hiletgo-usb-logic-analyzer-device-with-emi-ferrite-ring-usb](https://www.desertcart.com.eg/products/135070800-hiletgo-usb-logic-analyzer-device-with-emi-ferrite-ring-usb)

---

*Product available on Desertcart Egypt*
*Store origin: EG*
*Last updated: 2026-02-21*